|Year : 2010 | Volume
| Issue : 2 | Page : 190-195
Comparative study of RP-HPLC and UV spectrophotometric techniques for the simultaneous determination of amoxicillin and cloxacillin in capsules
Do T Giang, Vu D Hoang
Department of Analytical Chemistry and Toxicology, Hanoi University of Pharmacy 13-15 Le Thanh Tong, Hanoi, Vietnam
|Date of Web Publication||12-May-2010|
Vu D Hoang
Department of Analytical Chemistry and Toxicology, Hanoi University of Pharmacy 13-15 Le Thanh Tong, Hanoi
| Abstract|| |
Reversed-phase HPLC and UV spectrophotometric techniques using water as solvent have been developed and validated for the simultaneous determination of amoxicillin and cloxacillin in capsules. For both techniques, the linearity range of 60.0-140.0 ΅g/mL was studied. The spectrophotometric data show that non-derivative techniques, such as absorbance ratio and compensation, and ratio spectra first-order derivative could be successfully used for the co-assay of amoxicillin and cloxacillin. Based on the statistical comparison of spectrophotometric and chromatographic data, the interchangeability between HPLC and UV spectrophotometric techniques has been suggested for the routine analysis.
Keywords: Absorbance ratio, amoxicillin, capsules, cloxacillin, compensation, HPLC, UV derivative spectrophotometry
|How to cite this article:|
Giang DT, Hoang VD. Comparative study of RP-HPLC and UV spectrophotometric techniques for the simultaneous determination of amoxicillin and cloxacillin in capsules. J Young Pharmacists 2010;2:190-5
|How to cite this URL:|
Giang DT, Hoang VD. Comparative study of RP-HPLC and UV spectrophotometric techniques for the simultaneous determination of amoxicillin and cloxacillin in capsules. J Young Pharmacists [serial online] 2010 [cited 2014 Mar 8];2:190-5. Available from: http://www.jyoungpharm.in/text.asp?2010/2/2/190/63168
| Introduction|| |
Amoxicillin, formerly amoxycillin [Figure 1]a, is a moderate-spectrum beta-lactam antibiotic used to treat infections caused by penicillin-sensitive gram-positive bacteria as well as some gram-negative bacteria.  Amoxicillin is resistant to inactivation by gastric acid. It is usually the drug of choice because it is more rapidly and more completely absorbed than other beta-lactam antibiotics when orally administered. To overcome its sensitivity to destruction by beta-lactamases, amoxicillin has been co-administered with clavulanic acid, a potent betalactamase inhibitor  in pharmaceutical preparations.
Cloxacillin is a semisynthetic penicillin used as the sodium salt [Figure 1]b to treat staphylococcal infections due to penicillinase-positive organisms.  Unlike amoxicillin, this antibiotic is incompletely absorbed from the gastrointestinal tract, and absorption is reduced by the presence of food in the stomach. To produce a wider spectrum of activity, cloxacillin may be co-formulated with other antibacterials, in particular with amoxicillin (ratio 1:1, w/w) in capsules.
In the literature, the high performance liquid chromatography (HPLC) technique has been reviewed as a valuable tool for the analysis of antibiotics in formulated and unformulated samples.  As a result, this technique has been widely used for the simultaneous determination of penicillins such as amoxicillin and cloxacillin in pharmaceuticals, biological fluids, and tissues. ,,,,
On the other hand, amoxicillin was also spectrophotometrically analyzed without prior separation using UV derivative techniques in the combination with clavulanic acid ,,, or in antibiotic pharmaceutical mixtures. ,,,
Albeit HPLC is often an official method for the analysis of antibiotics, the need for other simple, reproducible and accurate analytical methods still exists. Thus, this study was carried out to compare HPLC and UV spectrophotometric techniques using water only as solvent for the simultaneous determination of amoxicillin and cloxacillin in their combined capsules.
| Experimental Details|| |
Apparatuses and conditions
A UNICAM UV 300 double beam spectrophotometer (Thermo Spectronic, USA) with a fixed slit width (1.5 nm) connected to an IBM computer loaded with Thermo Spectronic VISION32 software and 1 cm quartz cell were used for the registration and treatment of absorption spectra. For all solutions, zero-order spectra were recorded over the range from 200.0 to 300.0 nm against a blank (water) at Intelliscan mode to enhance the signal-to-noise ratio of absorbance peaks without extended scan duration with a ∆λ = 0.1 nm (i.e. 30-120 nm/min). To get the best signal-to-noise ratio and resolution, spectra and their corresponding derivative ones were further smoothed by using Savitzky - Golay filter (order 3, number of coefficients 125).
For the compensation technique, at any wavelength λ, the absorbance (A) of a mixture of two species X and Y (which do not interact with each other) is governed by the law of absorbance additivities, A m =A X +A y . If the absorption of Y is subtracted from A m , the absorption characteristics of the mixture gradually approach that of X as C Y increases. Finally, the absorption curve of mixture coincides with the absorption curve of X at the end-point, for which C Y used as subtrahend is exactly the concentration of Y in the mixture. For a pure substance, the absorbance ratio at two wavelengths is constant over a certain range of concentration (i.e. independent of concentration and whether another absorbing component is present). Thus, the identification of Y is based on this ratio i.e. the absorbance ratio of the mixture is equal to that of pure Y meaning the concentration of Y in the sample solution is equal to that of pure Y.
For the absorbance ratio technique, the principle is based on the linear relationship between the absorbancy ratio value of a binary mixture and the relative concentration of such a mixture. The quantification analysis of AMO and CLO in a binary mixture is performed using the following equations:
where Q=A/A iso
C 1 and C 2 : concentrations of AMO and CLA, respectively
A iso : absorbance at isosbestic point
a iso : absorptivity at isosbestic point ()
a 1 : slope of regression equation ()
a 2 : slope of regression equation ()
b 1 and b 2 : intercept values of these regression equations
A: absorbance of mixture solution measured at the mixture's maximum wavelength
10 3 : conversion factor of concentration unit from mg/mL to μg/mL.
For the UV derivative techniques, first-order derivative and ratio spectra first-order derivative spectra were evaluated for possible simultaneous determination of AMO and CLO.
High performance liquid chromatogram (HPLC) analysis was performed on an Agilent 1100 Series Diode-Array-Detector chromatograph (Agilent Technologies, USA) at ambient temperature. An Apollo C 18 (150 Χ 4.6 nm, 5μm) was used. All solutions were filtered through a 0.45 μm membrane filter before injection into the chromatograph. All solvents were filtered through a 0.45 μm Millipore filter and degassed in an ultrasonic bath.
Reagents and standard solutions
Amoxicillin trihydrate, AMO (98.4%), cloxacillin sodium, CLO (98.3%), and all excipients were kindly provided by Pharbaco Central Pharmaceutical Joint-Stock Company (Vietnam). De-ionized doubly distilled water was used throughout. All reagents were of analytical grade.
Stock solutions of AMO and CLO (1000 μg/mL) were prepared in water. Standard series of solutions were prepared in 25 mL calibrated flasks by using the same stock solutions and all dilutions freshly made.
FACLACIN 2, brand-name drug commercially available in the domestic market (produced by Pharbaco Central Pharmaceutical Joint-Stock Company, Vietnam, containing 250 mg AMO and 250 mg CLO per capsule) was studied.
The contents of 10 capsules were accurately weighed and finely powdered in a mortar. A mass corresponding to one-tenth of a capsule was transferred to a 100 mL calibrated flask containing about 30 mL water, well shaken, and ultrasonicated for 15 min. After the dissolution process, the solution was filtrated in a 100 mL calibrated flask through Whatman grade No. 42 filter paper. The residue was washed three times with 10 mL water and the volume completed to 100 mL with water. The resulting solution was further diluted to 1:5 in 25 mL calibrated flask with the same solvent for UV spectrophotometric measurement.
| Results and Discussion|| |
Non-derivative UV spectrophotometric techniques
[Figure 2](a) shows the zero-order UV absorption spectra in water indicating that the two spectra of AMO and CLO overlapped greatly in the wavelength region 200.0-300.0 nm. AMO exhibited a maximum at 271.8 nm, whilst CLO showed only a band of gradually reducing absorbance from 210.0 nm at 280.0 nm. Because the additivity of absorbances was not obeyed as clearly shown in [Figure 2]b, the use of bivariate equation technique,  a.k.a Vierordt's method,  selecting extrema of the two compounds was avoided due to error potentially acquired.
However, AMO and CLA could be still simultaneously determined using zero-order spectra such as compensation and absorbance ratio techniques. Unlike previously cited technique using derivative spectra, , the compensation technique in our study involves a comparison of several zero-order spectra (mixture - standard) using different concentrations of a standard solution as subtrahends. To determine the absorbance ratio at two selective wavelengths, a series of solutions containing different concentrations of pure drugs, above and below that presented in the binary mixture solution were analyzed [Table 1]. It is very important to mention that at the two wavelengths selected 250.2 and 252.5 nm, the mixture spectrum and addition spectrum coincided completely.
On the other hand, binary mixtures containing AMO and CLA could be analyzed by the absorbance ratio technique  as described above. For this technique, the wavelengths for isosbestic point and mixture's maximum absorbance were 259.0 and 271.8 nm, respectively [Figure 3].
Derivative UV spectrophotometric techniques
[Figure 4](a) displays the first derivative spectra of these pure drugs revealing that there existed two zero-crossing points at 258.7 and 271.6 nm for AMO, whereas no zero-crossing point observed for CLO. Only 258.7 nm was chosen for the determination of CLO due to its derivative amplitudes at this wavelength proportional to the concentration range studied 60.0-140.0 μg/mL.
In contrast, [Figure 4](b) shows two zero-crossing points at 269.6 and 281.4 nm for CLO. The second-order derivative values for AMO at these wavelengths, however, were not adequate for the correct determination of AMO from corresponding regression equations. This means AMO could not be determined using the UV first-order derivative technique.
[Figure 5](a, b) present the first-order derivatives of zero-spectra used for the determination of AMO and CLO. To optimize this technique, the influence of divisor standard concentration was investigated with the concentration ranges for Lambert-Beer's law compliance 60.0-140.0 μg/mL. A standard spectrum of 60.0 μg/mL was considered as suitable for the determination of both drugs. The determination of each component was based on the proportionality of its concentrations to relevant first-order derivative amplitudes at a suitable wavelength. 258.0 and 266.2 nm were chosen as working wavelengths for analyzing CLO and AMO, respectively.
In our study, the reversed-phase HPLC technique was developed to provide a specific procedure for the analysis of binary mixture containing AMO and CLO. The optimization of HPLC analysis was as follows. AMO and CLO were chromatographically analyzed by isocratic elution with a flow rate of 1.0 mL/min. The mobile phase composition was 0.01 M KH 2 PO 4 - methanol (45:55, v/v). Injection volume was 20 μl and detection wavelength was 225.0 nm for both compounds. Under our chromatographic conditions, the retention times were found to be 1.57 and 4.00 min for AMO and CLO, respectively [Figure 6]. The chromatographic parameters such as resolution (Rs = 15.68), peak asymmetry (AF < 1.10), and plate number (960 / 15 cm) were satisfactory for both compounds obviously confirming the suitability of our HPLC technique.
Method validation and application
The validity and suitability of the proposed UV spectrophotometric techniques were assessed by accuracy, precision, and linearity. For studying the accuracy, the proposed techniques were applied to simultaneously determine AMO and CLO in a synthetic mixture containing both the analytes and excipients which reproduced exactly the manufacturer's formula. The amount of analyte recovered was expressed as average percent recovery with the upper and lower limits of standard deviation. The average percent recoveries obtained were 98.93 ± 0.95 and 101.04 ± 0.87% for AMO and CLO, respectively, indicating the techniques' good accuracy and no marked interference by common excipients in the capsule studied. The within-run precision (repeatability) of these techniques was evaluated by analyzing six replicates of the synthetic mixture a day. The intermediate precision of these techniques was also evaluated with this mixture being analyzed during six consecutive days. The mean of relative standard deviation (%RSD = [S/X]Χ100, where S is standard deviation and X is mean of sample analyzed) were calculated. The low %RSD values (< 1.0%) of intra-day and inter-day variations indicate the proposed techniques' good precision.
By analogy, the HPLC technique also showed good accuracy (99.86 ± 0.69 and 100.15 ± 0.71% recovered for AMO and CLO, respectively) and precision (%RSD < 0.5%) with the linearity ranges 60.0-140.0 mg/mL for both drugs.
The calibration graphs for HPLC and UV derivative spectrophotometric determination are summarized in [Table 2].
The proposed techniques were successfully applied to the simultaneous determination of AMO and CLO in their combined capsules. The spectrophotometric results were statistically compared with those obtained by the HPLC technique [Table 3]. It is seen that at 95% confidence level, there was no significant difference between the accuracy (evaluated by the Student's t-value) and precision (evaluated by the variance ratio F-value) between the spectrophotometric techniques and HPLC.
| Conclusion|| |
In conclusion, the chromatographic separation of AMO and CLO in our study was characterized with good accuracy and precision. Except first-order derivative UV spectrophotometry, non-derivative techniques (i.e. absorbance ratio and compensation) and ratio spectra first-order derivative technique could be used for the simultaneous determination of AMO and CLO in their combined capsules. These UV spectrophotometric techniques using water as solvent were simple, reproducible, and accurate. Moreover, these techniques were statistically compared to chromatographic data suggesting possible interchangeability between UV spectrophotometric techniques and HPLC in the routine analysis.
| Acknowledgement|| |
The authors would like to thank Ms. Ha Thanh Hoa for technical help.
| References|| |
|1.||Sean CS. Martindale: The complete drug reference. 35 th ed. CD-ROM. "Amoxicillin" and "Cloxacillin" monographs. Pharmaceutical Press; 2004. |
|2.||Todd PA, Benfield P. Amoxicillin/clavulanic acid: An update of its antibacterial activity, pharmacokinetic properties and therapeutic use. Drugs 1990;39:264-307. |
|3.||Shalini J. HPLC separation of antibiotics present in formulated and unformulated samples. J Pharmaceut Biomed Anal 2002;28:795-809. |
|4.||Lambert KS, Birthe MR, Joe OB, Lily K. Simultaneous determination of six penicillins in cows' raw milk by a multiresidue high-performance liquid chromatographic method. J Chromatogr B Biomed Sci Appl 1997;694:383-91. |
|5.||Lambert KS, Lena KS, Tina E, Helga H. Simultaneous determination of seven penicillins in muscle, liver and kidney tissues from cattle and pigs by a multiresidue high-performance liquid chromatographic method. J Chromatogr B Biomed Sci Appl 1999;734:307-18. |
|6.||Samanidou VF, Evaggelopoulou EN, Papadoyannis IN. Development of a validated HPLC method for the determination of four penicillin antibiotics in pharmaceuticals and human biological fluids. J Sep Sci 2006;29:1550-60. |
|7.||Samanidou VF, Nisyriou SA, Papadoyannis IN. Development and validation of an HPLC method for the determination of penicillin antibiotics residues in bovine muscle according to the European Union Decision 2002/657/EC. J Sep Sci 2007;30:3193-201. |
|8.||Hsieh SH, Huang HY, Lee S. Determination of eight penicillin antibiotics in pharmaceuticals, milk and porcine tissues by nano-liquid chromatography. J Chromatogr A 2009;1216:7186-94. |
|9.||Ezzat MA, Mohammad AA, Mohamed EM, Nashaat AK. Spectrophotometric determination of amoxycillin and clavulanic acid in pharmaceutical preparation. Talanta 1989;36:683-5. |
|10.||Ewa BG. Determination of amoxycillin and clavulanic acid in some pharmaceutical preparations by derivative spectrophotometry. Mikrochim Acta 2001;136:31-4. |
|11.||Muρoz de la Peρa, A, Espinosa-Mansilla A, Acedo Valenzuela MI, Goicoechea HC, Olivieri AC. Comparative study of net analyte signal-based methods and partial least squares for the simultaneous determination of amoxycillin and clavulanic acid by stopped-flow kinetic analysis. Anal Chim Acta 2002;463:75-88. |
|12.||Abdil Φ, AyΊenur K. Comparative study of electrospray mass spectrometry and first derivative method and validation by HPLC method. J Food Drug Anal 2007;15:118-25. |
|13.||Murillo JA, Rodrνguez J, Lemus JM, Alaρσn A. Determination of amoxicillin and cephalexin in mixtures by second-derivative spectrophotometry. Analyst 1990;115:1117-9. |
|14.||Ezzat MA. Short communication: spectrophotometric determination of amoxycillin and dicloxacillin in binary mixtures and in capsules. J Pharmaceut Biomed Anal 1991;9:187-90. |
|15.||Akgόl Y, Mevlόt E. Second derivative spectrophotometric determination of amoxycillin in oral suspensions. Anal Lett 1991;24:2033-41. |
|16.||Gupta AK, Kaskhedikar SG. Derivative spectrophotometric estimation of amoxycillin and bromhexine hydrochloride in tablets. Asian J Chem 2003;15:977-80. |
|17.||Glenn AL. The importance of extinction ratios in the spectrophotometric analysis of mixtures of two known absorbing substances. J Pharm Pharmacol 1960;12:595-608. |
|18.||Erdal D. A comparative study of the ratio spectra derivative spectrophotometry, Vierordt's method and high-performance liquid chromatography applied to the simultaneous analysis of caffeine and paracetamol in tablets. J Pharmaceut Biomed Anal 1999;21:723-30. |
|19.||Wahbi AA, El-Yazbi FA, Barary MH, Sabri SM. Derivative spectrophotometric analysis of two-component mixtures using a compensation technique. Analyst 1992;117:785-9. |
|20.||Altun ML, Erk N. The rapid quantitative analysis of phenprobamate and acetaminophen by RP-LC and compensation technique. J Pharmaceut Biomed Anal 2001;25:85-92. |
|21.||Enk N, Kartal M. Comparison of high - performance liquid chromatography and absorbance ratio methods for the determination of hydrochlorothiazide and lisinopril in pharmaceutical formulations. Anal Lett 1999;32:1131-41. |
[Figure 1], [Figure 2], [Figure 3], [Figure 4], [Figure 5], [Figure 6]
[Table 1], [Table 2], [Table 3]